EXFO OTDR MANUAL PDF
EXFO OTDR 2 • User guide • Download PDF for free and without registration!. Get EXFO OTDR 2 Owner’s manual. Get all EXFO manuals!. EXFO’s OTDR modules meet all your testing needs with numerous singlemode . Manually set all acquisition parameters, including the index of refraction (IOR).
|Genre:||Health and Food|
|Published (Last):||16 July 2017|
|PDF File Size:||20.79 Mb|
|ePub File Size:||8.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
From the Save As dialog box, select a folder or create one to save your file.
EXFO OTDR 2 User Manual
The loss specified does not indicate the true loss of the event. The comments can be seen in the tooltip.
The application will automatically switch to the selected view when you perform the next acquisitions or open already existing files. Page Page 8 Managing Trace Files from the OTDR Test Application Once you have acquired traces, or when you want manial work with them after an acquisition, you will need to save, open, rename, and delete trace files.
You can also display various relative power readings. Enter the desired information. Contents 9 Creating and Generating Reports The OTDR then detects and analyzes the backscattered signals. It is potentially harmful in direct intrabeam viewing.
Keep the unit out of direct sunlight. Manhal the Increment window, select the Auto Increment check box corresponding to the identifier you want to increment. If you want to apply the test configuration information to the current acquisition, proceed as follows: The information of the selected element event or fiber section are displayed on the left of the linear view.
Contents Contents Certification Information If you select more than one marker, they will move as a block. The OTDR calculates distances as follows: Fully featured, entry-level, dedicated OTDR with tablet-inspired design perfect for frontline singlemode fiber installers.
If you want to keep manuao auto-settings activated once an acquisition is done, proceed as follows: Tap the Value column corresponding to the desired identifier.
Page Analyzing Traces and Events Using a Reference Trace Using a Reference Trace Setting a measurement as a reference allows you to test fibers and compare them to a reference trace that was previously acquired and analyzed. The information you enter is saved for the currently open trace file or the next acquisition depending on your choice. The Revert to Factory Settings button resets all the values in the Report tab.
To add favorites to a list, you must be logged in to your account. From the Spans list, select the desired span type. By default, all traces are generated in a report, but you can also generate all the traces the current file contains. Page 90 Page 91 6 Analyzing the Results Manually You can move markers and zoom in on or out of any event or trace segment to measure splice loss, fiber section attenuation, reflectance, and optical return loss.
Select the event on which you want to modify or delete a comment. To set the launch and receive settings for the next acquisition: Detected events are distinct even if they cover more than one point on the trace.
Warranty Liability Liability EXFO shall not be liable for damages resulting from the use of the product, nor shall be responsible for any failure in the performance of other items to which the product is connected or the operation of any system of which the product may be a part.
OTDR and iOLM software solutions| EXFO
Page 54 Testing Fibers After analysis, events appear in the events table. From the wavelength list, tap the wavelength value not the checkbox corresponding to the wavelength that you want to monitor. A loss value is specified for positive events. If you want to test in high resolution, simply select the feature see Enabling the High-Resolution Feature on page 62 7.
In this case, the only button available to start an acquisition is the Start Real Time button. Tell us manaul it.