HACH SC100 MANUAL PDF
View and Download Hach Sc E instrument manual online. Analysis System. Sc E Measuring Instruments pdf manual download. Hach Sc Controller User Manual – Download as PDF File .pdf), Text File .txt) or read online. Hach Sc Controller User Manual . Hach Company, – .. 37 W with 25 W sensor/network card load 24 VDCpowered sc model: 24 VDC.
|Published (Last):||3 June 2006|
|PDF File Size:||4.47 Mb|
|ePub File Size:||9.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
Appendix A Modbus Protocol A. Section 5 System Startup 5.
Index Numerics 90 Degree Majual Problem resolution and fuse replacement should only be performed by qualified service personnel. Depends on sample Clean the sensor characteristics. The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk.
HACH SC E INSTRUMENT MANUAL Pdf Download.
Page 14 Installation 3. Page 21 Installation 3. Figure 22 Front of the Controller 1 sc 5 2 6 3 7 4 1.
Pay attention to all danger and caution statements. Wet Verification System Startup 5.
Measurement Xfer Transfer 4. Vis spectrophotometer for hach and hach lange water analysis methods 24 pages.
Hach-Lange SC User Manual | 58 pages
Section 11 Limited Warranty In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase manjal, subject to the pro-rated schedule above, excluding original shipping and handling charges.
Your manual failed to upload Table 1 Turbidimeter Specifications Nephelometric light scatter at 90 degrees relative to the incident light beam. To hold the output until released: Pipe vertical or horizontal 5.
Failed fuses are an indication xc100 an equipment problem could exist.
The warranty period for instruments is 24 months. Page 72 Visit http: Disconnect and remove the existing wires between the quick connect and terminal block J5, see Figure 21 on page 4. Instrument Setup Operation 4.
Supplied mounting hardware is shown in Figure 6, Figure 7, and Figure 8. Gasket, Neoprene, panel mount, Cat.
Hach-Lange SC 100 User Manual
Please contact the Hach Service Center serving your location. Wire gauge less than 18 AWG is not recommended. Operation During calibration, analog outputs can remain active, held, or transferred to a preset mA value. The time is available only in hour military format. Troubleshooting Section 7 Troubleshooting Wear cotton gloves to protect your hands and to avoid fingerprints on the bulb. Pan head screw, M6 x 1,0 x mm 4Cat.
What’s missing? Tell us about it.
Pass the cable through conduit and a conduit hub or sc1000 strain relief fitting and an available access hole in the controller enclosure. Remove the quick connect fitting and wires and install the threaded plug on the opening to maintain the environmental rating.
Rinse with deionized water before calibrating. The data logs store the measurement data at selected intervals.